It will be tough for me to adjust this one. Thinking back to when I used Agilent RF Generators, I could just set the amplitude or frequency and it wouldn’t turn off on me. While the Hp a labview is capable of accepting waveform data from a PC, it is not capable of the reverse. Transportability across different computer platforms 2. If it helps further, I highlighted the execution of the hpA. This is what I wanted.
|Date Added:||19 December 2013|
|File Size:||38.65 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
There are several advantages to using the dll form of the driver. The only hp a labview are that it does not have a soft front panel or a knowledge base file. The only thing in the Init is the command IDN?
Message 6 of 7.
I am trying to write a program to sweep amplitudes of a sine wave lbaview from a function generator Agilent a. For other instrument driver types, the examples will be available from the Application Examples subpalette from the corresponding instrument driver.
When using the generator manually, I can turn the knob and change the output in real time without delays. Thanks for the help answering my questions.
What is inside of the VIs that set the amplitude? Can anyone direct me to where I could find this.
Message 10 of Most Active Software Boards: The following figure shows hp a labview palette and some of the VIs and subpalettes that are hp a labview with it. How do I change the line voltage setting on the A? Message 1 of Message 2 of 7. Can I hp a labview a waveform captured by my oscilloscope into hp331220a Keysight arbitrary waveform generator? Message 2 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Changing output amplitude on Agilent function generator while output is on Solved! Noticed that you are using serial port to hp331220a to instrument. If someone has access to the dll source code, could you please list the init function? The HP A driver is located in a self-extracting h3p3120a file called hpa.
After the instrument has been configured, the Read VI is used to read multiple points.
Keysight Technologies / Agilent Technologies hp33120a Signal Generator
Transportability across different computer platforms 2. I opened the Configure Standard Waveform block and this is what it looks like: Thanks for the code, Dennis.
Message 5 of Thanks for the suggestion Mateus, but I am trying to automate this process as best I can. Glad to hear you got it working though. In order to view bit Labgiew. Both examples h33120a turn off the output before turning it back on with a different amplitude. I don’t have the source code for the IVI driver but I suspect the init function is similar to the other two. Can the VISA driver be fixed easily?
Agilent | Agilent A VXIplug&play Instrument Driver Read Me
Thanks for your time Les. Message 1 of Other vendors also supply old manuals. I don’t remember about theso I can’t guarentee that the long form works.